Using Different HPLC Column Chemistries To Maximise Selectivity For Method Development
Sep 17 2018 Read 1755 Times
The purpose of HPLC method development is to determine conditions that provide adequate separation of analytes of interest in a reasonable time. Obtaining suitable selectivity should be the first step in this process, before final optimisation of the separation. Column chemistry (or more accurately the mechanisms of interaction between the stationary phase and the analyte that determine retention and separation) is one of the most powerful parameters that can be used to maximise selectivity. Sometimes phases other than C18 can be more helpful! In this Knowledgebase article, we discuss different column chemistries and their mechanisms of interaction to see how they can be a powerful ally for rapid LC method development.
Selectivity (α) may be defined as the ratio of retention factors (k) for two adjacent eluting analyte peaks in a chromatogram as depicted in Figure 1.
The separation selectivity may be affected by many parameters that include mobile phase conditions, stationary phase chemistry and the instrument setup. Some parameters are known to have a larger impact on selectivity, whilst others can have a smaller influence. Column stationary phase chemistry is known to have a significant effect upon selectivity .
HPLC is an important technique across many different application areas for separating and quantifying analytes of interest
within mixtures. This requires the analytes of interest to be sufficiently resolved from adjacent peaks in the chromatogram. From a practical point of view, resolution can be defined by Equation 1, where
Rs = resolution, N = number of theoretical plates, α = selectivity and k = retention factor.
Changing selectivity, efficiency or retention can affect resolution. However, if we represent the equation graphically (Figure 2), theory clearly indicates that selectivity is the most powerful term with respect to maximising resolution within this mathematical expression.
For method development purposes, exploring column chemistry and mechanisms of interaction to maximise selectivity is therefore helpful to resolve analytes of interest.
Reversed-Phase LC Mechanisms of Interactions
The most common column chemistry used across many application areas is C18.
The C18 chemistry typically provides good performance and excellent column lifetimes / reproducibility under various conditions.
It is also seen as broadly applicable for many analyte classes. The dominant retention and separation mechanism with alkyl chain based stationary phases is hydrophobic interactions. Depending upon the stationary phase chemistry, other mechanisms may be possible: should the stationary phase contain an aromatic ring or substituted aromatic ring (e.g. C18-AR or C18-pentaflurorophenyl (PFP)) and the analyte has regions of aromaticity, then pi-pi interactions may be possible. If polarisable functionality exists within the stationary phase chemistry (e.g. C18-PFP or CN-ES ), then dipole-dipole interactions between the stationary phase and analyte may be achieved. Mechanisms such as hydrogen bonding (from the silica surface and polar moieties, e.g. polar embedded groups such as C18-Amide or carbamate) and shape selectivity (C18-PFP and C18) are also known. It is possible to determine the weighting of such mechanisms on each stationary phase using published defined protocols and tests . This helps the analyst understand which mechanisms may operate with different stationary phases. Table 1 lists some different stationary phase chemistries and their main mechanisms of interaction. More stars indicate a higher weighting for that particular mechanism with the stationary phase.
Using complementary stationary phases during LC method development increases the potential interactions between analyte and stationary phase, therefore maximising selectivity and resolution. A simple and powerful way to optimise column selectivity is to screen a sample on a selection of different stationary phases using a generic set of gradient conditions. Figure 3 shows the chromatograms obtained using this approach for each of the stationary phases in Table 1, for the analysis of paracetamol (as the parent compound) and 9 related substances (spiked at 0.5% w/w).
The coloured boxes highlight retention and selectivity differences with the different column chemistries. The green box highlights how peak 4 may elute before, after, or completely coelute with the parent compound peak 1. Depending upon the mechanisms of interaction the selectivity is quite different. The red box highlights peak pair 7 & 8. Whilst generally well separated on all phases, the elution order is observed to reverse with the polar embedded ACE C18-Amide and ACE CN-ES stationary phases. Peaks 9 & 10 range from partial coelution to complete coelution and fully separated depending upon the stationary phase. Interestingly, it can be noted from these data that the ACE C18-Amide and ACE CN-ES provide complete separation of all 10 components compared to the other phases. The different mechanisms of interaction for these 2 phases provide suitably different selectivity to obtain complete resolution. One of the key points from these data is that the ACE C18 chemistry would not have separated the components using these gradient conditions. Further work would have been required, involving more time to determine a suitable separation. Using the power of stationary phase chemistry to maximise selectivity, there are 2 options where complete resolution of the 10 components is achieved, without the need for further method development.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Articles - Capillary Electrophoresis-Mass Spectrometry for Micro-Metabolomics - Imaged Capillary Iso-Electric Focussing: Background, Status, and Perspectives - High-Throughp...
View all digital editions
Jun 02 2019 Atlanta, GA, USA
Jun 16 2019 Milan, Italy
Jul 07 2019 Baltimore, MD, USA
Jul 10 2019 Baltimore, MD, USA
Aug 25 2019 San Diego, CA, USA