Automated Logical Dilution for Ion Chromatography
Jul 07 2010 Read 1381 Times
To get accurate results in chromatography, the determined sample peak areas must be within the limits of the calibration. Moreover, high ionic concentrations can overload the column. Automated inline dilution of samples with Metrohm’s complete system of hard- and software for IC meets both challenges, thus providing accurate determinations of anions and cations.
A simple step-by-step procedure
In order to perform automated dilutions, the 850 Professional IC, the 858 Professional Sample Processor and the MagIC™ Net software are additionally equipped with a magnetic stirrer and an 800 Dosino. The latter aspirates a defined volume of undiluted sample and transfers it to the mixing vessel. Subsequently the dilution medium, generally ultrapure water, is dosed into the mixing vessel. After intensive stirring, the peristaltic pump of the sample changer transfers the sample to the injection valve of the IC.
Automatic verification for reliable results
Alternatively, the sample can be injected directly. After the quantitation of the sample’s constituents, the software verifies if the peak areas lie within the calibration range. If this is the case, the next sample will be analyzed. If not, the software, after calculating the dilution factor, will re-analyze the sample after applying the appropriate dilution.
Summary: Whether the sample is diluted or directly injected, automated intelligent dilution ensures that peak areas are always within the limits of calibration.
For further information please contact Metrohm UK Ltd by Emailing your requests to email@example.com
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Articles - Enhanced Sample Preparation - Identifying Inherent Contamination in Deep Well Microplates - How to Determine Extra Column Dispersion and Extra Column Volume - Th...
View all digital editions
Jul 29 2018 Washington DC, USA
Aug 02 2018 Barcelona, Spain
Aug 06 2018 Berlin, Germany
Aug 26 2018 Florence, Italy
Sep 05 2018 Chiba, Japan