How to Determine Extra Column Dispersion and Extra Column Volume
Jun 13 2018
Advanced Chromatography Technologies Ltd.
on behalf of Advanced Chromatography Technologies
The width of a chromatographic peak is dependent on various dispersion processes operating both inside and outside the column. Extra column volume (ECV) is an important source of extra column dispersion (ECD) and can have significant detrimental effects, resulting in losses in column efficiency and therefore peak resolution. This Knowledgebase article outlines how ECV and ECD can be easily measured.
When an analyte is injected onto an LC instrument, it migrates through the system and column in a discrete band. The measured width of the band is affected by dispersive processes occurring within the column packed bed (intra-column) and within the LC system components (extra-column). The total observed peak dispersion (σ2tot), defined as the variance of a Gaussian shaped peak, can be expressed as follows:
σ 2 tot=σ 2 col+σ 2 ext
Where σ 2 col and σ 2 ext are the contributions from intra-column dispersion and extra-column dispersion (ECD) respectively.
Whilst intra-column dispersion is linked to the column, ECD has two contributors that may be optimised by the chromatographer. Extra column volume (ECV) is comprised of the internal volumes of various system components including tubing, fittings, autosampler and detector flow cell and is a major contributor to ECD. If ECV is large, then significant band broadening may occur, resulting in wider peaks and a loss in efficiency and resolution. Time- related factors such as detector sampling rate also affect the observed peak width. It is important to ensure detector settings are set to acquire enough data points to accurately describe the peak. This Knowledgebase article explains how ECV and ECD can be easily determined for any LC system.
The influence of ECV is generally insignificant for large volume columns (e.g. 150 x 4.6 mm). However, for smaller volume columns (e.g. 50 x 2.1 mm), the effects of ECV are more pronounced. Applications utilising small columns packed with small particles therefore require LC systems with highly optimised ECV, i.e. UHPLC systems; although it is possible to substantially reduce the ECV in many standard LC systems. Figure 1 shows the same isocratic separation run on a 75 x 3.0 mm column on systems with low and high ECV. Even for this moderate dimension column, the impact of ECD on resolution and efficiency is clearly evident and would be more pronounced on a smaller i.d. column.
If excessive ECD is suspected as the cause of poor method performance, ECV should be reduced where possible or the method transferred to a more optimised system. Alternatively, the method can be translated to a column with a larger i.d. to minimise the impact of ECV. As mentioned, data capture rate should also be optimised, particularly for very fast separations, or those using small volume columns. An insufficient data capture rate can result in poor description of chromatographic peaks, resulting in artificially broad peaks and loss in efficiency.
It is useful to experimentally determine both ECV and ECD for LC systems in order to ensure that the appropriate column format is utilised. The ECV and ECD can easily be determined using the following procedure.
ECD can have a large negative impact on column performance, particularly when performing separations on smaller volume columns. ECV is a major contributor to ECD. Experimental determination of the ECV and ECD of any LC system allows the chromatographer to select a suitable column format for their application, in order to maximise efficiency and achieve optimum chromatography.
Chromatography Today - Buyers' Guide 2022
In This Edition Modern & Practocal Applications - Accelerating ADC Development with Mass Spectrometry - Implementing High-Resolution Ion Mobility into Peptide Mapping Workflows Chromatogr...
View all digital editions
Jun 04 2023 Houston, TX, USA
Plastics Recycling World Expo Europe
Jun 14 2023 Essen, Germany
Jun 18 2023 Dusseldorf, Germany
Jul 04 2023 Cardiff, UK
Jul 11 2023 Lagos, Nigeria