What is HPLC-UV?
Nov 08 2018 Read 5749 Times
Alongside spectroscopy, chromatography is probably one of the main instrumental analytical techniques used in modern laboratories. There are many different types of chromatography – but they all follow the same essential principles. One of the main forms of chromatography used in laboratories is HPLC, and HPLC is commonly combined with spectroscopy in the form of a UV detector – giving rise to the powerful technique HPLC-UV.
HPLC – liquid chromatography performs separations
Chromatography originally started out as a technique performed at atmospheric pressure and relying on gravity – but as technology progressed it was found that using higher pressures – nowadays we use pressures between 50 and 300 bar - allowed faster and better separations. Hence, the name high pressure liquid chromatography that over time has morphed into high-performance liquid chromatography as analysts realised that the technique performs at an impressive level.
Chromatography is the science of separation – it separates molecules by pumping, or allowing, a mobile phase to flow through a stationary phase. The mobile phase carries the sample through a column or other device that holds the stationary phase. The stationary phase often is made from silica molecules or polymer. The stationary phase can be either polar (normal phase) or non-polar (reversed phase) depending on the sample properties and the separation required
It is the degree of interaction between sample molecules in the mobile phase and the stationary phase that determines how the sample is separated into its components. By using different columns and solvent combinations, it is possible to analyse many different sample types – method development is discussed in the article, Using Different HPLC Column Chemistries To Maximise Selectivity For Method Development.
Spectroscopy and chromatography – a good fit
A chromatography system is in essence very simple – a method of introducing the sample, stationary and mobiles phases to separate molecules and a means to detect the separated molecules. It is the ability to perform post-separation analysis on a sample that gives chromatography it strength as an analytical method.
There are many different detectors used in chromatography – flame-ionisation, thermal conductivity and electron capture are some of the commonly used detectors. But perhaps the most commonly used detector in liquid chromatography is ultra violet absorption - giving rise to the method HPLC-UV.
A UV detector uses a molecules ability to absorb ultraviolet light – detectors can use a single wavelength or variable wavelengths which can be more sensitive. By passing UV light through the individual components of an eluting sample mix and measuring the amount of UV light absorbed by each component, you can determine the individual quantities of each component.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Articles Modern & Practical Applications - LC-MS Analysis of Therapeutic Oligonucleotide and Related Products - Expanding the Boundaries of Light Scattering for Macromolec...
View all digital editions
Mar 08 2021 Virtual Event
May 09 2021 Fort Worth, Tx, USA
Jun 01 2021 Essen, Germany
Jun 06 2021 Philadelphia, PA, USA
Jun 06 2021 Baltimore, MD, USA