Capillary Electrophoresis coupling to Mass Spectrometry (CE-MS), an advanced technique orthogonal to LC-MS for high resolution separation and accurate molecule identification - Martin Greiner
Dec 17 2010 Comments 0
Free to read
This article has been unlocked and is ready to read.
Capillary Electrophoresis (CE) coupled with Mass Spectrometry (MS) has been under investigation for the past couple of decades1. Combining the high separation power of CE with MS promises to yield more sensitive detection and higher information content on the analytes than current methods such as standard UV detection. Since those early days, there have been many improvements and technology advancements on CE and MS instruments, as well as the interfacing devices. Modern Time of Flight (TOF) MS instruments and robust commercial interfaces provide increasingly sensitive routine analysis with accurate mass data.
The basic approach to CE-MS is to use Electrospray Ionization (ESI) techniques. The advantage to this system is that typically ionized or polar compounds separated by the CE under low flow conditions (low nl/min range) can then be efficiently transferred from the liquid phase into gas phase ions required by MS instruments. In addition, ESI does not decompose fragile molecules, like proteins. Routine CE-MS requires the use of volatile
buffers or low concentration salts. This avoids contamination of MS instruments and the suppression of analyte ions by salts competing for entrance to the MS. Separation by CE and ionization of molecules in the ESI source are both based on analyte chemistry and require optimization for best results. It is critical to choose the correct solvents and pH ranges for solubility, to allow for optimal CE mobility and for high ionization efficiency during desolvation. This may call for different chemistries for the separation buffers and for a sheath liquid controlling the ESI processes.
In this issue... Contents - Foreword by Trevor Hopkins Articles - The Emergence of Multiple Attribute Methodology (MAM) Workflows - Stationary Phase Selection for Achiral Separation ...
View all digital editions
Jan 24 2018 Cardiff, UK
Feb 05 2018 Dubai, UAE
Mar 18 2018 Dubai, UAE
Mar 18 2018 New Orleans, LA, USA
Mar 27 2018 Guangzhou, China