Separation and purification of samples using the new Kromasil 300 Å diC4
Sep 27 2017 Read 653 Times
Kromasil 300 Å diC4 spherical silica stationary phase is the latest Kromasil phase launched for the separation and purification of large peptides and proteins. This new stationary phase is the alternative to large pore, C4 derivatized materials used for the analysis and isolation compounds including those in complex biomolecule samples.
Preparative chromatographers focus on the isolation of key compounds from substance characterization or toxicology studies to process development work for clinical studies and production of APIs. When smaller quantities of material are required, scientists and researchers carry out their runs in slurry packed columns with 10-50 mm in diameter and for larger quantities users purify material by packing their own columns. Variations in stationary phase chemistry are exploited for best selectivity and overload to achieve conditions to meet the required purification goals. Kromasil stationary phases are recognized worldwide for being the workhorses in the purification of low molecular weight compounds, small peptides as well as large biomolecules.
In particular, the Kromasil 300 Å line of products, originally consisting of C18, C8 and C4 derivatizations for reversed-phase HPLC, has now been expanded to include the Kromasil 300 Å diC4 stationary phase. This new stationary phase differs in characteristics to the classic C4, giving HPLC users new opportunities to succeed in the removal of unwanted impurities from peptide and protein crudes. While similar in carbon content and packing density to the Kromasil 300 Å C4, the new Kromasil 300 Å diC4 contains a ligand with two C4. For chemistry characteristics see Figure 1.
Evaluation of Kromasil 300 Å C4 and diC4 materials were carried out for a given biosample, see separation in Figure 2. As seen in the figure, due to the impurities’ characteristics in this sample, there was better resolution with the diC4 ligand, where the retention of key impurities was lower than the main peak. This benefit was especially important for achieving certain purity and simultaneously bring down impurity concentrations to a required level (example < 0.1% impurities). Further evaluations were carried out using the well-known Tanaka test (refer to Figure 3). This test is a tool to determine hydrophobicity, steric selectivity, silanol activity and ion exchange capacity. The study concluded that the diC4 had higher silanol activity, exhibited lower retention capacity had better ion exchange capacity at neutral pH. The Kromasil 300 Å diC4 phase showed to be more advantageous when the sample had impurities that were tougher to elute, had strong adsorption.
As a member of the 300 Å family, this new phase has all the benefits, such as high available surface, and excellent mechanical and chemical stability, from the Kromasil Classic platform.
For more information please click here.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Articles - Enhanced Sample Preparation - Identifying Inherent Contamination in Deep Well Microplates - How to Determine Extra Column Dispersion and Extra Column Volume - Th...
View all digital editions
Jul 29 2018 Washington DC, USA
Aug 02 2018 Barcelona, Spain
Aug 06 2018 Berlin, Germany
Aug 26 2018 Florence, Italy
Sep 05 2018 Chiba, Japan