Optimised HPLC Analysis of Monoclonal Antibodies (mAbs)
Nov 23 2016 Read 966 Times
Monoclonal antibodies (mAbs) are complex macromolecules that are developed as biotherapeutics themselves, or as targeting vehicles for delivery of toxic payloads (antibody drug conjugates). As complex macromolecules, they require multipronged approaches to analysis and characterisation, whether under development or to support production QC. Analysis of the intact mAb, can provide first level assessment of size, post-translational modifications, and heterogeneity. Such questions can be readily addressed by reversed-phase chromatography (RPC) coupled to the analytical power of mass spectrometry. RPC of proteins, particularly as large as IgG, requires the use of a large pore chromatographic matrix, a hydrophobic stationary phase, and appropriate chromatographic conditions – mobile phase and temperature. The typical phase for RP protein chromatography is a short alkyl chain, which provides adequate retention. By way of RP mobile phase conditions, however, mAbs generally require an elevated temperature for efficient chromatography. It has also been reported that the chromatography can be further enhanced by the inclusion of a protic solvent as a component of the organic modifier (the strong solvent). In addition to optimising the chromatographic conditions, new developments in particle technology have resulted in gains in chromatographic performance. These technology developments have been applied to both traditional porous particles and new core-shell (or superficially porous) particle design. For porous particles, the advancements have come in smaller particle size and chemical stability. For core-shell particles, the technological advancement is in the particle architecture itself - limiting the chromatographic separation process to the thin outer porous layer. Additionally, the synthetic route to the core-shell particle may yield a narrower particle size distribution, which in turn can affect chromatographic packing such that improved column efficiency is attained.
The figures compare the chromatographic results of two such modern columns designed for intact protein analysis. In Figure 1, the sample is the mAb adalimumab (aka Humira®), while for Figure 2, the sample is a mAb standard from MilliporeSigma (MSQC4). Both new columns were run on the same day with the same sample, on the same UHPLC instrument. A common method was used for both columns, at a temperature optimum of 75°C. It is clear that the core-shell column provides considerably narrower peak width (and thus better sensitivity), and as a result of the superior performance, also better resolves minor components of the sample. The enhanced performance of the core-shell column will be best realised in combination with a modern UHPLC instrument of minimal system dispersion. Please click here for additional information.
|column||as indicated, 10 cm x 2.1 mm I.D.|
|mobile phase A||80:20, (water, 0.1% TFA):(acetonitrile, 0.1% TFA)|
|mobile phase B||50:50, (water, 0.1% TFA):(acetonitrile, 0.1% TFA)|
|flow rate||0.2 mL/min|
|gradient||45 to 57% B in 12 min|
|det||UV, 215 nm|
|injection||0.3 µL Humira or 1.0 µL SigmaMAb|
|sample||mAb diluted in 20 mM Tricine/KOH, pH 7, to 1 g/L|
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Issue... Articles - Industrial Applications of Offline and Online Comprehensive Gas Chromatography - Improving Workflow in Group-Type GC×GC Analysis of Petrochemicals - Identifi ca...
View all digital editions
Jun 03 2018 San Diego, CA, USA
Jun 11 2018 Frankfurt, Germany
Jun 19 2018 Busan, Korea
Jul 08 2018 Baltimore, MD, USA
Jul 29 2018 Washington DC, USA