GCxGC with parallel MS and FID detection for improved analysis of fragrances
Sep 24 2019 Read 1064 Times
Gas chromatography Mass Spectrometry (GC-MS) is broadly employed for the analysis of allergens and aroma compounds in fragrances. However, the analytical challenge is not trivial. The complexity of essential oils and perfumes typically requires multiple analyses with different stationary phases to achieve suitable resolution for all targets.
Comprehensive two-dimensional gas chromatography (GC×GC) couples two columns in a single analysis to achieve superior separation power. Peaks are distributed across a 2D separation space based on different physicochemical properties, enhancing resolution and significantly reducing co-elution. These advantages make GC×GC-MS increasingly more widespread in the fragrance industry for accurate and time-effective targeted analysis as well as untargeted profiling.
In this application note Dr. Daniela Peroni demonstrates the feasibility of thermally-modulated GC×GC with parallel MS/FID detection to combine enhanced identification capability and robust, reliable quantification in fragrance analysis.
Instrument: 7890B GC equipped with FID and 5975C MSD with Triple-Axis detector (Agilent).
Modulator: Cryogen-free loop-type thermal modulator (ZOEX) with Optimode module (SRA Instruments).
Software: MassHunter for acquisition, GC Image for data visualization and processing.
Set-up: Outlet flow split to MSD/FID with a microfluidic 3-ports splitter (Trajan) and uncoated capillaries.
Results and discussion
In this configuration with parallel detection (Fig. 1) the choice of the correct splitter is of the upmost importance. Since GC×GC peaks are very sharp, band broadening and tailing should be minimised to preserve resolution. With this is mind, we selected a low-dead-volume planar microfluidic device. The performance of this set-up is assessed with a mixture of linear alkanes. Peak shape in terms of width and symmetry is satisfactory for both signals (Fig. 2). In this work we use identical capillaries to connect the splitter to the detectors, if needed the MSD/FID ratio can be adjusted by using restrictions with different, appropriate dimensions. This is useful, for instance, to increase sensitivity at the FID.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Articles - Elimination of the Sample Solvent Effect when Analysing Water Solutions of Basic Peptides by HILIC - Method Development and Validation of Simultaneous Determinatio...
View all digital editions
Jun 07 2020 Taipei, Taiwan
Jun 20 2020 San Diego, CA, USA
Aug 29 2020 Rio de Janeiro, Brazil
Sep 21 2020 Budapest, Hungary
Sep 27 2020 Saint-Malo, France