8 Common Gas Chromatography Mistakes
Aug 18 2014 Read 20634 Times
If you're new to Gas Chromatography and looking for some helpful tips, here are 8 common mistakes you'll want to avoid:
1) Incorrect gas flow rates to a flame ionization detector
If the flow rates are not correct the FID can give an incorrect response for a sample. The gas flow rate to the detector should be optimised according to the manufacturer’s recommendations. A typical ratio is 10:1 (air:hydrogen) producing rates of the order 300 mL/min air to 30 mL/min hydrogen. It is good practice to heat the detector to its operating temperature and to allow it to equilibrate before turning on the detector gases. If the detector does not light, check the flow rates; don’t rely on the instrument settings. If the detector still does not light, the air supply can be reduced to 150 mL/min to light the flame and then be slowly increased to the correct value.
2) Heating a column without any carrier gas flow
This is a simple way to damage a column. Always check that the carrier gas is on and flowing before the heat to the Gas Chromatograph (GC) oven is turned on. The carrier gas should not be turned off until the oven has cooled down.
3) Running out of gas
Making sure that there is sufficient gas for the GC runs. It is poor practice and can be costly to run out of gas - read the article, 5 Gas Management Tips for GC, for some useful advice on gas management.
4) Flooding the injection liner with sample
If the sample volume injected into the GC is too large for the liner, backflash can occur that can result in spurious chromatogram peaks and repeatability problems. Injection volumes should take into account the temperature of the inlet and the type of solvent being used. There can be a considerable difference in the gas volumes generated by different solvents when a sample vaporizes after injection.
5) Leaky septum
A septum is a consumable and should be inspected and changed regularly. It can leak after repeated use leading to pressure problems. Using a damaged septum can also lead to blockages if the septum breaks up. Make sure the correct type of septum is used for the application and that it is installed correctly.
6) Column not conditioned
New columns should always be conditioned before use. This not only improves the column efficiency, but also helps to prolong the life of the column. For more tips on making a GC column last longer take a look at 5 Ways to Extend a Gas Chromatography Column’s Life.
7) Using the wrong syringe
There are many different types and sizes of GC syringes. Consider the application and sample size before selecting the syringe type and volume. For example, trying to use a large volume syringe for a small sample volume can easily lead to problems including flooding of the inlet.
8) Preventative maintenance not carried out
This applies to all analytical instrumentation. Preventative maintenance is usually the lowest cost option compared to replacing broken parts. A simple daily and weekly schedule can identify problems before they happen.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Articles - Elimination of the Sample Solvent Effect when Analysing Water Solutions of Basic Peptides by HILIC - Method Development and Validation of Simultaneous Determinatio...
View all digital editions
Jun 07 2020 Taipei, Taiwan
Jun 20 2020 San Diego, CA, USA
Aug 29 2020 Rio de Janeiro, Brazil
Sep 21 2020 Budapest, Hungary
Sep 27 2020 Saint-Malo, France